TY - JOUR
T1 - Optimized projection ordering schemes for Compton camera reconstruction using OSEM
JF - Journal of Nuclear Medicine
JO - J Nucl Med
SP - 1479
LP - 1479
VL - 50
IS - supplement 2
AU - Kim, Soo Mee
AU - Lee, Jae Sung
AU - Lee, Chun Sik
AU - Kim, Chan Hyeong
AU - Lee, Myung Chul
AU - Lee, Dong Soo
AU - Lee, Soo-Jin
Y1 - 2009/05/01
UR - http://jnm.snmjournals.org/content/50/supplement_2/1479.abstract
N2 - 1479 Objectives Although the OSEM algorithm does not converge to the true ML solution, it is known to provide a good solution if the projections that constitute each subset are reasonably balanced. Here we investigate three different projection ordering schemes of OSEM for Compton cameras, which can efficiently produce subsets in a balanced way. Methods The Compton scattered data can be subdivided into subsets by a combination of scatting angles (SA) and detected positions (DP). To construct subsets which are balanced so that voxel activity contributes equally to any subset, the data were first arranged by three ordering schemes; the random ordering scheme (ROS), the multilevel ordering scheme (MOS), and the weighted-distance ordering scheme (WOS). The arranged data were then split into N subsets. To compare three ordering schemes, we calculated the coefficient of variation (CV) of the angular and positional differences and the percentage errors (PE). Results All ordering schemes showed the order-of-magnitude acceleration over the standard EM, and their computation times were about the same. The MOS and WOS provided the maximum mean-difference in case of SA- and DP-based arrangement, respectively. In contrast, the WOS exhibited minimum CVs for both SA- and DP-based arrangements. Conclusions The WOS that yielded the best results with the lowest PEs by providing larger and more uniform angular and positional differences in the SA- and DP-arrangements would be the optimal choice for Compton camera reconstruction using OSEM.
ER -